Got a second customer coming in today with a E507XB series network analyzer that does not turn on. Looks like it’s a common problem within the model series.
If the unit turns on without the CMOS battery but it doesn’t turn on the 2nd time after the CMOS was cleared, you have a very expensive problem which likely I’m the only person who can solve it because it’s months of effort tracing the circuit and the timings inside. I chasing regretted chasing down the rabbit hole and spent more labor than 3 good unit’s costs for the few grands I’ve charged, but I might be able to recoup the labor in the future as more of the E507XB fails the same way.
UNLESS it’s a SC815E (there are 2 revisions that uses different motherboard), it’s NOT the motherboard or the power supply. I’ve seen some faulty SC815E doing the same on other model series, but not the VP22s (I replaced 4~5 different VP22s, they all do the same thing). It’s some good timing issues that’s hard to pinpoint to a specific module (you can replace everything on the digital side and it still doesn’t work) which I had to design, build and test a special circuit to correct it.
If you have a unit E5060 series or E5070 series VNA (Vector Network Analyzer) that doesn’t boot, I’m likely to have the exact experience fixing it. Of course I’m open to solving other problems with the analyzer as well. I offer free eval (no fix, no fee). Please email me at firstname.lastname@example.org or call me at 949-682-8145.
8 total views