Architecturally, the DSO6000A series shares common designs with 54830 series oscilloscope and 54640 series oscilloscopes. I noticed the Acq board numbers for my DSO6052A start with 54672 and the my M/DSO6104A starts with 54684. Looks like the 5467X means 500Mhz and 5468X means 1Ghz while the X is the number of analog channels.
This is ‘confirmed’ by a slip up in the documentation (user guide) which they forgot to update the model number in their vector graphics:
I think I’m getting a hang of Agilent’s oscilloscope’s hardware to do deep level board repairs as I have various model to compare. I used If you need yours repaired, please reach me at 949-682-8145.