As time goes by, it seems like there’s quite a few small problems with the very well engineered 54830 series oscilloscope. These are all manageable (not core) problems and I believe I got the details down.
Models supported: 54830B, 54830D, 54831B, 54831M, 54831D, 54832B, 54832D
‘D’ means it’s a Mixed-Signal oscilloscope (comes with logic analyzer)
‘M’ is the military versions that’s exactly the same as ‘B’. The ‘M’ batch is for sure shipped with Windows 98.
I also support upgrading any Windows 98 based unit to Windows XP Pro.
I also have plenty of units and parts to get your problems fixed.
If it doesn’t power or boot, it might be a less severe problem than you think. Evaluation is always free.
Trade in for a guaranteed unit
Serviced units (Motherboard corrected, replaced mechanical HDD to SSD, replaced CD-ROM with DVD writer, comes with LS-120 drive, Windows XP Pro installed, 2 years repair/exchange* warranty):
- 54831M[$2999]: 600Mhz, 4GS/s
- 54831M modded to 54832B [$3499]: 1GHz, 4GS/s.
- 54832B [$3999]: 1GHz, 4GS/s
- 54832D [$4499]: 1GHz, 4GS/s (Mixed Signal Oscilloscope with Logic Analyzer)
If your decaying unit is frustrating you, you can swap your it with the worry-free, fully serviced units by paying a $1500 difference to enjoy all the upgrade and peace of mind.
* I’m likely to have a replacement unit for immediate exchange to get you up and running. In the unlikely event I run out of units, I’ll pay you $100/week delay if the repair takes more than 1 week, up to the full amount you’ve paid.
NOTE: This (swap) deal does not apply to units with a genuinely failed Acq board (like analog signal path/cal/trigger problems) or I confirmed that it has a failed power supply unit.
Parts cost (they are taken from fully working units, so there’s sacrifice):
- Corrected VP22 motherboard [$699]
[Subtract $200] if you trade in a defective VP22 (4 PCI slots).
[Subtract $300] if you trade in a working M880 (3 PCI slots).
- Old style interface card [$699]:
[Subtract $200] if you trade in your working GPIB combo card.
Old style interfaces works across the board. The combo only works on the latest acq boards.
Labor/Repair cost (all service guaranteed for 2 years). No cost if not successful:
- Front-end problem (cannot calibrate) [$599 for first channel + $199/channel]. Parts included.
- Cannot power up [$199]. Parts NOT included. Not charged if no actual repairs were made.
- Boot problems [$199]. Waived if parts were bought to solve it.
- Oscilloscope program hangs / BSOD [$199]. Waived if parts were bought to solve it.
- Replace motherboard [$199]. Parts NOT included.
Upgrade / service cost
- Upgrading from Windows 98 to Windows XP Pro [$499 for DIY, $699 for full service]: SSD included. This covers the self-calibration.
If your unit comes with licenses, you will need to provide the license number (can easily do it yourself) to enable them. Can ask Keysight to look it up for you.
- Modernize the unit [$299, $149 when combined with any repair/service/parts above]: migrate hard drive to SSD, replace CD-ROM with DVD drive, replace FDD with LS-120 drive (if applicable).
The prices are negotiable if more than one part/service is needed.
Call me at 650-804-5024. I’m located in Irvine, CA. You’ll be responsible for all shipping costs for anything that’s not my fault.
101 total views, 1 views today